RTD Calibrator is a exactitude measurement and source instrument, it can be use to calibrate the RTD* transmitter (include most impulse transmitter).
RTD Calibrator can measure or simulate 7 difference types of RTD (℃ or℉), and measure or simulate the Resistance. But it could not use to measurement or source at a same time.
|Range||0.00~3200.0Ω or -200.0~630.0℃|
|Display||Large 5 digit LCD|
|LCD Size||64×42 mm|
|Product Size||205 x 97 x 45 mm|
|Weight||~472 g (include battery)|
RTD Process Calibrator is a exactitude measurement and source instrument, it can be use to calibrate the RTD* transmitter (include most impulse transmitter).
Delivered with the multimeter:
2 pair of test lead and alligator clip, 6 * AAA 1.5V battery, user’s manual.
Periodically wipe the case with a damp cloth and detergent; do not use abrasives or solvents. Calibration
Calibrate your calibrator once a year to ensure that it performs according to its specifications.
Replacing the Battery
Please change the battery when the LCD indicates
Turn off the power of the Calibrator, When you change the battery, and screw off the breechblock on the battery cabinet cover, then take off it and instead the fresh AAA 1.5V battery.